- Open Access
The development of loop-mediated isothermal amplification targeting alpha-tubulin DNA for the rapid detection of Plasmodium vivax
© Dinzouna-Boutamba et al.; licensee BioMed Central Ltd. 2014
- Received: 7 March 2014
- Accepted: 26 June 2014
- Published: 30 June 2014
Malaria that is caused by Plasmodium vivax is the most widely distributed human malaria. Its recent resurgence in many parts of the world, including the Republic of Korea (ROK), emphasizes the importance of improved access to the early and accurate detection of P. vivax to reduce disease burden. In this study, a rapid and efficient loop-mediated isothermal amplification (LAMP)-based method was developed and validated using blood samples from malaria-suspected patients.
A LAMP assay targeting the α-tubulin gene for the detection of P. vivax was developed with six primers that recognize different regions of the target gene. The diagnostic performance of the α-tubulin LAMP assay was compared to three other tests: microscopic examinations, rapid diagnostic tests (RDTs), and nested polymerase chain reactions (PCRs) using 177 whole blood specimens obtained from ROK military personnel from May to December 2011.
The α-tubulin LAMP assay was highly sensitive with a detection limit of 100 copies of P. vivax α-tubulin gene per reaction within 50 min. It specifically amplified the target gene only from P. vivax. Validation of the α-tubulin LAMP assay showed that the assay had the highest sensitivity (P < 0.001 versus microscopy; P = 0.0023 versus RDT) when nested PCR was used as the gold standard and better agreement (concordance: 94.9%, kappa value: 0.865) with nested PCR than RDT and microscopy. A Receiver Operation Characteristics analysis showed that the diagnostic accuracy of the α-tubulin LAMP assay for vivax malaria was higher (Area Under Curve = 0.908) than RDT and microscopy.
This study showed that the P. vivax α-tubulin LAMP assay, which can be used to diagnose early infections of vivax malaria, is an alternative molecular diagnostic tool and a point-of-care test that may help to prevent transmission in endemic areas.
- Plasmodium vivax
- Immunochromatographic rapid diagnostic tests
- Polymerase chain reaction
- Loop-mediated isothermal amplification
Plasmodium vivax, which is a causative agent of human malaria, is the most widely distributed species, and 2.8 billion people are at risk for transmission in the world. Vivax malaria reemerged in 1993 in the Republic of Korea (ROK or South Korea), and its incidence rapidly increased to 4,142 by 2000. Plasmodium vivax malaria is commonly believed to be clinically benign and self-limiting[2, 3]. However, accumulating lines of evidence have shown that the impacts of P. vivax malaria with respect to economic and social burdens in endemic regions have been underestimated[4–6]. Moreover, the resurgence of vivax malaria in many parts of the world, including the ROK[7, 8], emphasizes the importance of improving access to reliable diagnostic methods that facilitate the early and accurate diagnosis of malaria, which is urgently required to facilitate disease management and control.
Microscopic examinations of Giemsa-stained thick and thin blood films, which are considered the gold standard for the diagnosis of malaria[10, 11], are recommended by the World Health Organization. Although this technique is highly specific, its sensitivity for the detection of P. vivax is lower than for Plasmodium falciparum due to the low parasitaemia of P. vivax. Furthermore, it is time consuming, labour-intensive, and requires technical expertise with respect to the interpretation of Giemsa-stained blood smears. Immunochromatographic rapid diagnostic tests (RDTs) that are based on the detection of histidine-rich protein-2 and/or lactate dehydrogenase[11, 13] provide rapid and straightforward field tests for the detection of P. vivax but are limited in terms of sensitivity and specificity[14, 15]. Accordingly, molecular diagnostic methods, such as polymerase chain reaction (PCR) and nested PCR, have been developed and used to improve P. vivax detection[16–21]. Although these assays have been shown to be highly effective for diagnosing malaria, they require laboratory equipment, trained personnel, and have long turnaround times, which limit their usefulness for routine diagnoses in the field.
Loop-mediated isothermal amplification (LAMP), which is a relatively straightforward and sensitive technique that is based on rapid DNA amplification under isothermal conditions, was recently developed to remove the need for sophisticated and expensive thermal cyclers. LAMP involves the specific amplification of target DNA by Bacillus stearothermophilus (Bst) DNA polymerase, allowing strand displacement DNA synthesis with a set of six oligonucleotides that recognize independent regions of the target gene. The use of six oligonucleotides improves the specificity and speed of the amplification and forms a loop-structured amplicon, which produces a typical ladder-pattern of multiple bands. A positive reaction is easily determined by eye as turbidity or fluorescence by the inclusion of fluorescent detection dyes, such as SYBR green or hydroxynaphthol under UV light[26, 27]. These features allow LAMP assays to be used to detect many pathogenic organisms, such as viruses, bacteria, fungi, parasites, and vivax malaria parasites[28–31].
The present study was undertaken to develop an α-tubulin targeting LAMP assay for the detection of P. vivax and to validate the assay using whole blood from suspected malaria patients. The sensitivity and specificity of the devised P. vivax α-tubulin LAMP assay were determined and compared with those of microscopy and RDTs with 18S ribosomal RNA (rRNA)-based nested PCR as gold standard. To validate the accuracies of the α-tubulin targeting LAMP assay, the performances of the tests examined were assessed using receiver operating characteristic (ROC)[32, 33].
This study was conducted at Armed Forces Hospitals that treat soldiers stationed near the DMZ, which separates the ROK from the Democratic People’s Republic of Korea (DPRK or North Korea), in the northern part of the Gyeonggi-do Province, in the northwest region of the ROK (between 37°–38° latitude and 127°–128° longitude). This is a high-risk area for malaria and where only P. vivax is transmitted. All enrolled soldiers had no history of travel to malaria-endemic areas and had never received a blood transfusion. Whole blood samples were collected by sequential sampling from 177 male ROK soldiers, who provided written informed consent, among all 189 ROK male soldiers who had been admitted to the Armed Forces Hospitals (from May to December 2011) with febrile illness (temperature ≥ 38°C) and were clinically suspected to have malaria. To detect P. vivax, approximately 1 mL of blood was collected by venipuncture into a vacutainer tube (Becton Dickinson, Franklin Lakes, NJ, USA) containing ethylenediaminetetraacetic acid (EDTA). Samples were transported on ice within 4 h to a laboratory (Department of Parasitology and Tropical Medicine, Kyungpook National University School of Medicine, Daegu, ROK) where they were stored at -70°C. As is required by the Declaration of Helsinki, donor confidentiality was maintained throughout, and the study was approved by the Ethics Committee of the Armed Forces Medical Command (AFMC-13-IRB-053, July 2011).
Microscopic examination and rapid diagnostic tests
These samples were assayed using standard diagnostic procedures, including direct microscopic examination of Giemsa-stained thick and thin blood films and RDT. The 177 blood samples from malaria-suspected patients were first examined with RDT and microscopy at the Armed Forces Hospitals. The results for microscopy and RDT were recorded by two different technicians. Then, each sample was blinded and transported to the laboratory (Department of Parasitology and Tropical Medicine, Kyungpook National University School of Medicine, Daegu) for further microscopic examination. A slide was considered negative if no asexual stages of Plasmodium spp. were found during examination of 100 fields. Parasite densities were assessed by counting against 200 leucocytes, and converting to parasites per microliter, assuming a standard leucocyte count of 8,000/μL. The immunochromatographic RDT (SD malaria Ag Pf/Pan, Standard Diagnostic, Inc., Hagal-Dong, Korea) detects the parasite antigen Plasmodium falciparum Histidine-rich protein-2 (PfHRP-2) specific to P. falciparum in one capture site and pan-Plasmodium lactate dehydrogenase (pan-pLDH) for all four Plasmodium species, in a separate capture line.
α-tubulin LAMP assay
Comparison of the results obtained from the four diagnostic methods
Diagnostic test results
No. of cases (%)
Comparison of the results obtained with the diagnostic methods on the 177 whole blood samples
Sensitivity (95% CId)
Specificity (95% CI)
% agreement with nested PCR (Kappa)
Analytical sensitivity and specificity of the α-tubulin LAMP assay
The α-tubulin gene was amplified by PCR from P. vivax with the F3 and B3 primers listed in Figure 1. The PCR products were purified with a Qiaquick gel extraction kit (QIAGEN, Inc., Valencia, CA, USA), and the eluted PCR products were cloned into a pGEM-T easy vector (Promega Corporation, Madison, WI, USA). The amplified DNA fragments were completely sequenced (Solgent Co., Ltd. Daejon, Korea) and confirmed to be identical to the sequence of the α-tubulin gene from the P. vivax Sal-1 strain. The concentrations of plasmid DNA were measured with a NanoDrop ND-1000 spectrophotometer (NanoDrop Products, Thermo Scientific Instruments, Inc., Wilmington, DE, USA), and the corresponding copy numbers were calculated. The plasmid containing the α-tubulin gene fragment was diluted with TE buffer (10 mM Tris–HCl, 1 mM EDTA) to final concentrations of 1, 10, 102, 103, or 104 copies of the gene per reaction. The specificities of the LAMP assays were evaluated with genomic DNA from P. falciparum 3D7, P. vivax Sal-1, Plasmodium malariae Uganda I/CDC, Plasmodium ovale curtisi, Plasmodium ovale wallikeri, Plasmodium knowlesi H strain, Toxoplasma gondii RH, Cryptosporidium parvum Iowa, and Babesia microti Munich strains. The DNAs of the Entamoeba histolytica HM1:IMSS and Giardia lamblia WB strains were kindly provided by Dr. Myeong Heon Shin and Dr. Soon-Jung Park (Yonsei University College of Medicine, Seoul, Korea).
Genomic DNA extraction from whole blood and nested PCR
The DNAs were prepared from 100 μL of whole blood with DNeasy tissue kits (QIAGEN, Inc.). One microlitre of extracted DNA that was dissolved in 20 μL of double-distilled water was used as a template for the nested PCR and LAMP assay. On the other hand, heat-treated lysates of whole blood samples were prepared as previously described with minor modification[36, 37]. Briefly, 1 ml of ice-cold 5 mM sodium phosphate buffer (pH 8.0) was added to 20 μl of whole blood sample, vortexed, and centrifuged at 12,000 × g for 5 min. The collected pellet was washed two times with 5 mM sodium phosphate buffer (pH 8.0), dissolved with 100 μl of distilled water, and boiled for 10 min. The samples were centrifuged again at 12,000 × g for 2 min, and the supernatants were collected. Four microliter aliquots of the supernatants collected were used as templates for LAMP. The nested PCR based on the 18S rDNA gene was performed as previously described in triplicate with TaKaRa LA Taq polymerase (Takara Bio Inc., Shiga, Japan) in a reaction volume of 20 μL with a thermal cycler (Perkin Elmer Cetus, PerkinElmer Inc., Waltham, MA, USA). Two primer sets for 18S rDNA gene, the universal primer (P1, 5′-ACGATCAGATACCGTCGTAATCTT-3′; P2, 5′-GAACCCAAAGACTTTGATTTCTCAT-3′) and the P. vivax specific primer (P1, 5′-ACGATCAGATACCGTCGTAATCTT-3′; V1, 5′-CAATCTAAGAATAAACTCCGA AGAGAAA-3′) that generated a 100-bp product, were employed per reaction. The primer sets that were used had The PCR products from each reaction were confirmed by DNA sequencing (Solgent Co., Ltd.). The amplified products were visualized on 1.5% agarose gels, stained with ethidium bromide, observed with a UV transilluminator, and independently recorded by two experienced investigators blinded against the other methods.
The test sensitivities and specificities were determined using nested PCR as gold standard. The percentage sensitivities and specificities with 95% confidence intervals (95% CIs) were calculated with MedCalc, version 7.0 (MedCalc Software bvba, Ostend, Belgium). Differences in the sensitivities and specificities were compared with McNemar’s normal approximation test with SAS 9.3 (SAS Institute, Inc., Cary, NC, USA). Differences between the areas under the receiver operating characteristic (AUC) were compared with a multiple logistic regression model and SAS 9.3 (SAS Institute, Inc.). The degree of agreement was determined using Kappa statistics in SAS 9.3 (SAS Institute Inc.). Statistical significance was accepted for p values less than 0.001.
Optimization of the P. vivax α-tubulin LAMP assay conditions
To develop a LAMP assay for the detection of P. vivax, a set of six primers targeting the α-tubulin gene was designed (Figure 1). The targeting region of the α-tubulin DNA in this LAMP assay showed relatively low sequence identity among Plasmodium spp. (P. knowlesi [GenBank accession no. XM_002258237]: 84.0% nucleotide identity; P. falciparum, [GenBank accession no. XM_001351490]: 77.6% nucleotide identity)(Figure 1A). The optimal temperature and time for the LAMP reaction were determined with a cloned α-tubulin gene fragment (106 copies per reaction) under isothermal conditions at temperatures of 60°C to 65°C for 120 min by monitoring turbidity. Although amplification targeting of the P. vivax α-tubulin gene was detected at all of the temperatures tested, a threshold value of absorbance (0.1) from the LAMP assay was reached most quickly at 64°C (data not shown). No nonspecific amplification was detected in the negative control (plasmid containing no insert) after at least 120 min of incubation. Thus, the subsequent LAMP reactions were conducted at 64°C for 90 min.
Analytical sensitivity and specificity of the P. vivax α-tubulin LAMP assay
Validation of the P. vivax α-tubulin LAMP assay
To validate the α-tubulin LAMP assays, the blood samples of 177 male ROK soldiers who were admitted with febrile illness and who were suspected of having malaria were subjected to α-tubulin LAMP and three available methods, including microscopic examinations, immunochromatographic RDTs and nested PCRs. To determine the accuracy of the tests for diagnosis of P. vivax malaria, nested PCR was used as a gold standard and the sensitivity, specificity, positive predictive value (PPV), and negative predictive value (NPV) were calculated for each test. Of the 177 male ROK soldiers, 96 (54%), 137 (77%), 128 (72%), and 137 (77%) were positive for microscopy, RDT, nested PCR, and α-tubulin LAMP, respectively. Microscopic examinations were positive for 96 specimens, resulting in a sensitivity of 75.0% (95% CI, 66.8%–81.7%) and a specificity of 100% (95% CI, 91.1%-100%) (Table 2). The median parasite density in the 96 male patients was 1,263 [103–5,177]/μL. All 96 positive specimens by microscopy were nested PCR positives. Thirty-two of the 81 negative specimens by microscopy were true positives by nested PCR and thus considered false negatives (Table 1). The results obtained by RDT were positive for 137 specimens, resulting in a sensitivity of 93.0% (95% CI, 87.9%–96.4%) and a specificity of 63.3% (95% CI, 49.2%—75.3%) (Table 2). Eighteen of the 137 positive (18/137, 13.1%) by RDT were negative by nested PCR for the detection of P. vivax and thus considered false positives. Nine of the 40 specimens (9/40, 22.5%) negative by RDT were true positive by nested PCR and thus considered false negatives (Table 1). All malaria-positive cases by RDT were identified as P. vivax infections.
The sensitivity of the α-tubulin LAMP assays was 100% (95% CI, 96.4%–100%), and its specificity was 81.6% (95% CI, 68.3%–90.2%) (Table 2). Eight that were positive by RDT but negative by nested PCR were positive by α-tubulin LAMP (Table 1). Of the 137 positives by α-tubulin LAMP, 128 specimens (136/137, 99.3%) were true positives, and thus 9 specimens were considered a false positive (Table 1). Ten of the 40 negative specimens by α-tubulin LAMP assays were only positive by RDT. On the other hand, α-tubulin LAMP assay conducted using heat-treated lysates produced the same results as those obtained using a commercial kit (DNeasy tissue kit) (data not shown). When nested PCR was used as the gold standard, the α-tubulin LAMP assays had the highest diagnostic sensitivity (P < 0.001 vs. microscopy; P = 0.0023 vs. RDT) and better agreement (concordance: 94.9%, kappa value: 0.865) with nested PCR than RDT and microscopy (Table 2). To compare the diagnostic accuracies of microscopy, RDT, and α-tubulin LAMP assays for the detection of P. vivax, the diagnostic accuracies were assessed by performing ROC analyses and measuring the AUCs. Among these three diagnostic tests when nested PCR was used as the gold standard, the α-tubulin LAMP assay had the highest AUC (0.908) (P = 0.4272 vs. microscopy; P = 0.0104 vs. RDT) among the three tests.
The resurgence of vivax malaria in many parts of the world emphasizes the need for a rapid, sensitive, and inexpensive diagnostic method. Furthermore, it needs to be easily performed and to produce readily interpretable results in the field of an endemic area. As one of the diagnostic methods of vivax malaria, microscopic examinations of blood samples are still widely used in many malaria-endemic areas, including the ROK. As shown in Table 2, microscopic examinations had the lowest sensitivity (75.0%) among the four tests in this study. Furthermore, the requirement of well-trained experts and the long turnaround time limit its usefulness in the field. Due to its convenience, RDT is currently and widely used as an alternative field diagnostic test. However, recent studies suggest that RDT had relatively lower sensitivity and specificity for P. vivax than P. falciparum[14, 15, 38]. The RDT in this study had high sensitivity relative to that of the microscopy, and had lower specificity and negative predictive values compared to those of α-tubulin LAMP assays by considering nested PCR as gold standard (Table 2). To solve these problems, molecular diagnostic tests, nested PCRs, and LAMP assays have been developed[11, 39]. In particular, P. vivax LAMP assays that are based on 18S rDNA, mitochondrial DNA, and Pvr64 have recently been developed and validated for the diagnosis of malaria[28–30, 40–44]. The sensitivities of nested PCRs and LAMP assays for P. falciparum malaria have been shown to be comparable or higher than RDTs, respectively. Thus, the aim of this study was to design and test the utility of a novel set of P. vivax-specific LAMP primers with different target DNA sequences in order to improve diagnostic performance and validate them as an alternative molecular diagnostic test. Although the tubulin gene of P. vivax is not species-specific, the β-tubulin gene sequence has previously been used successfully to amplify P. knowlesi DNA with a LAMP assay. The targeting region of the α-tubulin DNA in this LAMP assay showed relatively low sequence identity among Plasmodium spp. and its use has resulted in no cross reactivity with the other malaria species examined.
In the present study, the diagnostic performance of α-tubulin LAMP assays were compared to three currently available tests, including microscopic examinations, RDT, and nested PCRs using 177 whole blood specimens. When nested PCR was used as gold standard, 128 (72.3%) of the 177 specimens were positive for vivax malaria. PCR is still the most sensitive and specific method for vivax malaria among all of the currently available tests. However, expensive laboratory equipment, the need for a trained specialist, and time-consuming post-PCR procedures, such as agarose gel electrophoresis, limit the usefulness of nested PCR in the field when diagnostic equipment is minimal. However, a LAMP assay does not require a thermal cycler and can be performed with minimal laboratory facilities, such as a heating block or a water bath, and its end points can be determined by turbidity or color development by eye. Furthermore, the straightforward sample preparation for molecular diagnosis reduces the risk of cross-contamination, and thus the risks posed by false-positive results. LAMP assays that are based merely on heat-treated blood have previously been found to be as efficient at detecting malaria parasites as DNA that is extracted with a commercial kit[44, 48]. In this study, results of α-tubulin LAMP assay based on heat-treated samples were consistent with those of the LAMP assay using a commercial kit. Thus, the devised P. vivax LAMP assay could eliminate the need for DNA extraction without compromising sensitivity and reduce the time required to reach a diagnosis. The α-tubulin LAMP assay had the higher diagnostic accuracy (AUC = 0.908) than microscopy and RDT and showed better agreement with nested PCR (Table 2) and thus can be used in the diagnosis of early infections of vivax malaria as an alternative molecular diagnostic tool.
In endemic countries, including the ROK, vivax malaria is also transmitted by the transfusion of infected blood products[49, 50], and, as yet, no reliable approved laboratory test is available for the screening of donated blood. Although the LAMP assay was applied to whole blood samples of patients suspected to be infected, the LAMP assay devised here could also be used for blood screening for P. vivax. Furthermore, individuals with low parasite density can provide reservoirs for transmission, and thus an early and accurate diagnostic tool like the α-tubulin LAMP assay may help to prevent transmission in endemic areas.
This study describes about the development and validation of a LAMP assay for the detection of P. vivax DNA in clinical blood samples. In addition, the P. vivax α-tubulin LAMP assay developed here can be used to diagnose early infections of vivax malaria, is an alternative molecular diagnostic tool and a point-of-care test that may help to prevent transmission in endemic areas.
This research was supported by the Basic Science Research Program through the National Research Foundation of Korea funded by the Ministry of Education, Science and Technology (NRF-2010-0016679). We wish to express our appreciation to Dr. Kawai Satoru (Dokkyo Medical University) for providing P. knowlesi genomic DNA. We are indebted to Dr. Jung-Yeon Kim (The Korea Centers for Disease Control and Prevention) for providing the P. malariae and P. ovale DNA.
- Guerra CA, Howes RE, Patil AP, Gething PW, Van Boeckel TP, Temperley WH, Kabaria CW, Tatem AJ, Manh BH, Elyazar IR, Baird JK, Snow RW, Hay SI: The international limits and population at risk of Plasmodium vivax transmission in 2009. PLoS Negl Trop Dis. 2010, 4: e774-PubMed CentralView ArticlePubMedGoogle Scholar
- Galinski MR, Barnwell JW: Plasmodium vivax: who cares?. Malar J. 2008, 7 (Suppl 1): S9-PubMed CentralView ArticlePubMedGoogle Scholar
- Mueller I, Galinski MR, Baird JK, Carlton JM, Kochar DK, Alonso PL, del Portillo HA: Key gaps in the knowledge of Plasmodium vivax, a neglected human malaria parasite. Lancet Infect Dis. 2009, 9: 555-566.View ArticlePubMedGoogle Scholar
- Price RN, Tjitra E, Guerra CA, Yeung S, White NJ, Anstey NM: Vivax malaria: neglected and not benign. Am J Trop Med Hyg. 2007, 77: 79-87.PubMed CentralPubMedGoogle Scholar
- Kochar DK, Saxena V, Singh N, Kochar SK, Kumar SV, Das A: Plasmodium vivax malaria. Emerg Infect Dis. 2005, 11: 132-134.PubMed CentralView ArticlePubMedGoogle Scholar
- Genton B, D'Acremont V, Rare L, Baea K, Reeder JC, Alpers MP, Muller I: Plasmodium vivax and mixed infections are associated with severe malaria in children: a prospective cohort study from Papua New Guinea. PLoS Med. 2008, 5: e127-PubMed CentralView ArticlePubMedGoogle Scholar
- Chai IH, Lim GI, Yoon SN, Oh WI, Kim SJ, Chai JY: [Occurrence of tertian malaria in a male patient who has never been abroad]. Korean J Parasitol. 1994, 32: 195-200.View ArticlePubMedGoogle Scholar
- Park JW, Jun G, Yeom JS: Plasmodium vivax malaria: status in the Republic of Korea following reemergence. Korean J Parasitol. 2009, 47 (Suppl): S39-50.PubMed CentralView ArticlePubMedGoogle Scholar
- Reyburn H: New WHO guidelines for the treatment of malaria. BMJ. 2010, 340: c2637-View ArticlePubMedGoogle Scholar
- Bruce-Chwatt LJ: From Laveran's discovery to DNA probes: new trends in diagnosis of malaria. Lancet. 1987, 2: 1509-1511.View ArticlePubMedGoogle Scholar
- Moody A: Rapid diagnostic tests for malaria parasites. Clin Microbiol Rev. 2002, 15: 66-78.PubMed CentralView ArticlePubMedGoogle Scholar
- Barker RH: DNA probe diagnosis of parasitic infections. Exp Parasitol. 1990, 70: 494-499.View ArticlePubMedGoogle Scholar
- WHO: Malaria rapid diagnostic test performance: Results of WHO product testing of malaria RDTs Round 2 (2009). 2009, Geneva, Switzerland: WHOView ArticleGoogle Scholar
- Mens P, Spieker N, Omar S, Heijnen M, Schallig H, Kager PA: Is molecular biology the best alternative for diagnosis of malaria to microscopy? A comparison between microscopy, antigen detection and molecular tests in rural Kenya and urban Tanzania. Trop Med Int Health. 2007, 12: 238-244.PubMedGoogle Scholar
- Murray CK, Gasser RA, Magill AJ, Miller RS: Update on rapid diagnostic testing for malaria. Clin Microbiol Rev. 2008, 21: 97-110.PubMed CentralView ArticlePubMedGoogle Scholar
- Kimura M, Kaneko O, Liu Q, Zhou M, Kawamoto F, Wataya Y, Otani S, Yamaguchi Y, Tanabe K: Identification of the four species of human malaria parasites by nested PCR that targets variant sequences in the small subunit rRNA gene. Parasitol Int. 1997, 46: 91-95.View ArticleGoogle Scholar
- Perandin F, Manca N, Calderaro A, Piccolo G, Galati L, Ricci L, Medici MC, Arcangeletti MC, Snounou G, Dettori G, Chezzi C: Development of a real-time PCR assay for detection of Plasmodium falciparum, Plasmodium vivax, and Plasmodium ovale for routine clinical diagnosis. J Clin Microbiol. 2004, 42: 1214-1219.PubMed CentralView ArticlePubMedGoogle Scholar
- Rougemont M, Van Saanen M, Sahli R, Hinrikson HP, Bille J, Jaton K: Detection of four Plasmodium species in blood from humans by 18S rRNA gene subunit-based and species-specific real-time PCR assays. J Clin Microbiol. 2004, 42: 5636-5643.PubMed CentralView ArticlePubMedGoogle Scholar
- Singh B, Bobogare A, Cox-Singh J, Snounou G, Abdullah MS, Rahman HA: A genus- and species-specific nested polymerase chain reaction malaria detection assay for epidemiologic studies. Am J Trop Med Hyg. 1999, 60: 687-692.PubMedGoogle Scholar
- Snounou G, Viriyakosol S, Jarra W, Thaithong S, Brown KN: Identification of the four human malaria parasite species in field samples by the polymerase chain reaction and detection of a high prevalence of mixed infections. Mol Biochem Parasitol. 1993, 58: 283-292.View ArticlePubMedGoogle Scholar
- Snounou G, Viriyakosol S, Zhu XP, Jarra W, Pinheiro L, do Rosario VE, Thaithong S, Brown KN: High sensitivity of detection of human malaria parasites by the use of nested polymerase chain reaction. Mol Biochem Parasitol. 1993, 61: 315-320.View ArticlePubMedGoogle Scholar
- Hanscheid T, Grobusch MP: How useful is PCR in the diagnosis of malaria?. Trends Parasitol. 2002, 18: 395-398.View ArticlePubMedGoogle Scholar
- Notomi T, Okayama H, Masubuchi H, Yonekawa T, Watanabe K, Amino N, Hase T: Loop-mediated isothermal amplification of DNA. Nucleic Acids Res. 2000, 28: E63-PubMed CentralView ArticlePubMedGoogle Scholar
- Mori Y, Notomi T: Loop-mediated isothermal amplification (LAMP): a rapid, accurate, and cost-effective diagnostic method for infectious diseases. J Infect Chemother. 2009, 15: 62-69.View ArticlePubMedGoogle Scholar
- Mori Y, Nagamine K, Tomita N, Notomi T: Detection of loop-mediated isothermal amplification reaction by turbidity derived from magnesium pyrophosphate formation. Biochem Biophys Res Commun. 2001, 289: 150-154.View ArticlePubMedGoogle Scholar
- Goto M, Honda E, Ogura A, Nomoto A, Hanaki K: Colorimetric detection of loop-mediated isothermal amplification reaction by using hydroxy naphthol blue. Biotechniques. 2009, 46: 167-172.View ArticlePubMedGoogle Scholar
- Tomita N, Mori Y, Kanda H, Notomi T: Loop-mediated isothermal amplification (LAMP) of gene sequences and simple visual detection of products. Nat Protoc. 2008, 3: 877-882.View ArticlePubMedGoogle Scholar
- Han ET, Watanabe R, Sattabongkot J, Khuntirat B, Sirichaisinthop J, Iriko H, Jin L, Takeo S, Tsuboi T: Detection of four Plasmodium species by genus- and species-specific loop-mediated isothermal amplification for clinical diagnosis. J Clin Microbiol. 2007, 45: 2521-2528.PubMed CentralView ArticlePubMedGoogle Scholar
- Tao ZY, Zhou HY, Xia H, Xu S, Zhu HW, Culleton RL, Han ET, Lu F, Fang Q, Gu YP, Liu YB, Zhu GD, Wang WM, Li JL, Cao J, Gao Q: Adaptation of a visualized loop-mediated isothermal amplification technique for field detection of Plasmodium vivax infection. Parasit Vectors. 2011, 4: 115-PubMed CentralView ArticlePubMedGoogle Scholar
- Patel JC, Oberstaller J, Xayavong M, Narayanan J, DeBarry JD, Srinivasamoorthy G, Villegas L, Escalante AA, DaSilva A, Peterson DS, Barnwell JW, Kissinger JC, Udhayakumar V, Lucchi NW: Real-time loop-mediated isothermal amplification (RealAmp) for the species-specific identification of Plasmodium vivax. PLoS One. 2013, 8: e54986-PubMed CentralView ArticlePubMedGoogle Scholar
- Karanis P, Ongerth J: LAMP–a powerful and flexible tool for monitoring microbial pathogens. Trends Parasitol. 2009, 25: 498-499.View ArticlePubMedGoogle Scholar
- DeLong ER, DeLong DM, Clarke-Pearson DL: Comparing the areas under two or more correlated receiver operating characteristic curves: a nonparametric approach. Biometrics. 1988, 44: 837-845.View ArticlePubMedGoogle Scholar
- Campuzano-Zuluaga G, Hanscheid T, Grobusch MP: Automated haematology analysis to diagnose malaria. Malar J. 2010, 9: 346-PubMed CentralView ArticlePubMedGoogle Scholar
- Iwagami M, Fukumoto M, Hwang S-Y, Kim S-H, Kho W-G, Kano S: Population structure and transmission dynamics of Plasmodium vivax in the Republic of Korea based on microsatellite DNA analysis. PLoS Negl Trop Dis. 2012, 6: e1592-PubMed CentralView ArticlePubMedGoogle Scholar
- PrimerExplorer V4. [http://primerexplorer.jp/e/]
- Foley M, Ranford-Cartwright LC, Babiker HA: Rapid and simple method for isolating malaria DNA from fingerprick samples of blood. Mol Biochem Parasitol. 1992, 53: 241-244.View ArticlePubMedGoogle Scholar
- Henning L, Felger I, Beck H-P: Rapid DNA extraction for molecular epidemiological studies of malaria. Acta Trop. 1999, 72: 149-155.View ArticlePubMedGoogle Scholar
- Wongsrichanalai C, Barcus MJ, Muth S, Sutamihardja A, Wernsdorfer WH: A review of malaria diagnostic tools: microscopy and rapid diagnostic test (RDT). Am J Trop Med Hyg. 2007, 77: 119-127.PubMedGoogle Scholar
- Abdul-Ghani R, Al-Mekhlafi AM, Karanis P: Loop-mediated isothermal amplification (LAMP) for malarial parasites of humans: would it come to clinical reality as a point-of-care test?. Acta Trop. 2012, 122: 233-240.View ArticlePubMedGoogle Scholar
- Pöschl B, Waneesorn J, Thekisoe O, Chutipongvivate S, Panagiotis K: Comparative diagnosis of malaria infections by microscopy, nested PCR, and LAMP in northern Thailand. Am J Trop Med Hyg. 2010, 83: 56-PubMed CentralView ArticlePubMedGoogle Scholar
- Chen J-H, Lu F, Lim CS, Kim J-Y, Ahn H-J, Suh I-B, Takeo S, Tsuboi T, Sattabongkot J, Han E-T: Detection of Plasmodium vivax infection in the Republic of Korea by loop-mediated isothermal amplification (LAMP). Acta Trop. 2010, 113: 61-65.View ArticlePubMedGoogle Scholar
- Sirichaisinthop J, Buates S, Watanabe R, Han E-T, Suktawonjaroenpon W, Krasaesub S, Takeo S, Tsuboi T, Sattabongkot J: Short Report: Evaluation of Loop-Mediated Isothermal Amplification (LAMP) for Malaria Diagnosis in a Field Setting. Am J Trop Med Hyg. 2011, 85: 594-596.PubMed CentralView ArticlePubMedGoogle Scholar
- Polley SD, González IJ, Mohamed D, Daly R, Bowers K, Watson J, Mewse E, Armstrong M, Gray C, Perkins MD: Clinical evaluation of a loop-mediated amplification kit for diagnosis of imported malaria. J Infect Dis. 2013, 208: 637-644.PubMed CentralView ArticlePubMedGoogle Scholar
- Polley SD, Mori Y, Watson J, Perkins MD, González IJ, Notomi T, Chiodini PL, Sutherland CJ: Mitochondrial DNA targets increase sensitivity of malaria detection using loop-mediated isothermal amplification. J Clin Microbiol. 2010, 48: 2866-2871.PubMed CentralView ArticlePubMedGoogle Scholar
- Lee PW, Ji DD, Liu CT, Rampao HS, Do Rosario VE, Lin IF, Shaio MF: Application of loop-mediated isothermal amplification for malaria diagnosis during a follow-up study in Sao Tome. Malar J. 2012, 11: 408-PubMed CentralView ArticlePubMedGoogle Scholar
- Iseki H, Kawai S, Takahashi N, Hirai M, Tanabe K, Yokoyama N, Igarashi I: Evaluation of a loop-mediated isothermal amplification method as a tool for diagnosis of infection by the zoonotic simian malaria parasite Plasmodium knowlesi. J Clin Microbiol. 2010, 48: 2509-2514.PubMed CentralView ArticlePubMedGoogle Scholar
- WHO: Malaria Diagnosis: New Perspectives. 2000, Geneva, Switzerland: WHOGoogle Scholar
- Poon LL, Wong BW, Ma EH, Chan KH, Chow LM, Abeyewickreme W, Tangpukdee N, Yuen KY, Guan Y, Looareesuwan S: Sensitive and inexpensive molecular test for falciparum malaria: detecting Plasmodium falciparum DNA directly from heat-treated blood by loop-mediated isothermal amplification. Clin Chem. 2006, 52: 303-306.View ArticlePubMedGoogle Scholar
- Mungai M, Tegtmeier G, Chamberland M, Parise M: Transfusion-transmitted malaria in the United States from 1963 through 1999. N Engl J Med. 2001, 344: 1973-1978.View ArticlePubMedGoogle Scholar
- Oh JS, Kim JS, Lee CH, Nam DH, Kim SH, Park DW, Lee CK, Lim CS, Park GH: Evaluation of a malaria antibody enzyme immunoassay for use in blood screening. Mem Inst Oswaldo Cruz. 2008, 103: 75-78.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.