- Open Access
A semi-automated multiplex high-throughput assay for measuring IgG antibodies against Plasmodium falciparum erythrocyte membrane protein 1 (PfEMP1) domains in small volumes of plasma
© Cham et al; licensee BioMed Central Ltd. 2008
Received: 05 March 2008
Accepted: 12 June 2008
Published: 12 June 2008
The level of antibodies against PfEMP1 is routinely quantified by the conventional microtitre enzyme-linked immunosorbent assay (ELISA). However, ELISA only measures one analyte at a time and requires a relatively large plasma volume if the complete antibody profile of the sample is to be obtained. Furthermore, assay-to-assay variation and the problem of storage of antigen can influence ELISA results. The bead-based assay described here uses the BioPlex100 (BioRad, Hercules, CA, USA) system which can quantify multiple antibodies simultaneously in a small plasma volume.
A total of twenty nine PfEMP1 domains were PCR amplified from 3D7 genomic DNA, expressed in the Baculovirus system and purified by metal-affinity chromatography. The antibody reactivity level to the recombinant PfEMP1 proteins in human hyper-immune plasma was measured by ELISA. In parallel, these recombinant PfEMP1 proteins were covalently coupled onto beads each having its own unique detection signal and the human hyper-immune plasma reactivity was detected for each individual protein using a BioPlex100 system. Protein-coupled beads were analysed at two time points seven months apart, before and after lyophilization and the results compared to determine the effect of storage and lyophilization respectively on the beads. Multiplexed protein-coupled beads from twenty eight unique bead populations were evaluated on the BioPlex100 system against pooled human hyper-immune plasma before and after lyophilization.
The bead-based assay was sensitive, accurate and reproducible. Four recombinant PfEMP1 proteins C17, D5, D9 and D12, selected on the basis that they showed a spread of median fluorescent intensity (MFI) values from low to high when analysed by the bead-based assay were analysed by ELISA and the results from both analyses were highly correlated. The Spearman's rank correlation coefficients (Rho) were ≥ 0.86, (P < 0.0001) for all comparisons. Bead-based assays gave similar results regardless of whether they were performed on individual beads or on multiplexed beads; lyophilization had no impact on the assay performance. Spearman's rank correlation coefficients (Rho) were ≥ 0.97, (P < 0.0001) for all comparisons. Importantly, the reactivity of protein-coupled non-lyophilized beads decreased with long term storage at 4°C in the dark.
Using this lyophilized multiplex assay, antibody reactivity levels to twenty eight different recombinant PfEMP1 proteins were simultaneously measured using a single microliter of plasma. Thus, the assay reported here provides a useful tool for rapid and efficient quantification of antibody reactivity against PfEMP1 variants in human plasma.
The hope of developing a vaccine against malaria is based on evidence that clinical immunity to the disease is developed through repeated exposures over several years to the pathogen . Several studies suggest that protective immunity to malaria develop partly through the acquisition of a wide repertoire of specific antibodies directed against the polymorphic antigen target, Plasmodium falciparum erythrocyte membrane protein 1 (PfEMP1) [2, 3].
To date, anti-PfEMP1 antibody levels in human plasma samples have been measured using enzyme-linked immunosorbent assay (ELISA). As P. falciparum malaria predominantly affects individuals of young age, studies of malaria immunity rely on plasma samples from infants and toddlers. This creates a limitation in using ELISA as obtainable plasma volumes from these target groups are relatively small. In addition ELISA is time consuming and labor intensive. Recent technological advances have resulted in the development of high-throughput multiplex methods which enable the simultaneous detection of antibodies to multiple analytes in human plasma samples.
Vignali  described the use of the Luminex100 system, a bench-top flow cytometer equipped with two low power laser beams and capable of performing 100 discrete assays simultaneously in a single well. Each bead set is impregnated with a unique ratio of red-to-infrared dyes. When excited, each bead set emits its own unique detection signal that can be resolved by the instrument. Molecules covalently coupled to the beads, such as recombinant PfEMP1 proteins, can be detected by the use of a biotinylated secondary antibody with phycoerythrin-conjugated streptavidin used as a reporter.
Several studies have reported the use of multiplex assays to measure cytokine levels in samples , antibody levels to protein antigens  and antibodies to multiple malaria vaccine candidate antigens .
The assay reported here for evaluating the antibody profile of human plasma samples is based on a multiplex of twenty eight recombinant PfEMP1 protein coupled beads, each bead population with its own unique detection signal. The assay, requires one microliter of plasma sample for measuring antibodies to all twenty eight recombinant PfEMP1 proteins, is reproducible, gives results comparable to ELISA and is high-throughput. Importantly, the coupled beads remained stable after lyophilization and storage at -80°C.
Materials and methods
1-ethyl-3-[3dimethylaminopropyl] carbodiimide hydrochloride (EDC) and N-hydroxysulfosuccinimide (Sulfo_NHS) were purchased from Pierce Biotechnology (Rockford, IL). 2-[N-morpholino] ethanesulfonic acid (MES), Tween-20, bovine serum albumin (BSA) sodium azide, biotinylated anti-human IgG, biotinylated anti-V5 antibody and phycoerythrin conjugated streptavidin were purchased from Sigma-Aldrich, USA.
The hyper-immune plasma pool was made up of plasma from ten individuals from a malaria endemic area of Liberia. Twenty samples from Danes who have never had malaria were used to make up the naïve pool. Sixty individual plasma samples, twenty each from people living in the three Tanzanian villages Mgome, Ubiri and Magamba with high, moderate and low malaria transmission , respectively were also analysed.
Protein expression was as described previously [9, 10]. Briefly, primer pairs designed to contain restriction enzyme sites (See additional file 1) were used to amplify Cysteine-rich inter-domain regions (CIDR) and Duffy binding-like (DBL) domains from 3D7 genomic DNA. The digested PCR products were cloned into the Baculovirus vector, pAcGP67-A (BD Bioscience), which was designed to contain the V5 epitope upstream of a histidine tag in the C-terminal end of the construct. The identity of the cloned fragments was verified by sequencing. Linearized Bakpak6 Baculovirus DNA (BD Biosciences Clontech) was co-transfected with pAcGP67-A into Sf9 insect cells for generation of recombinant virus particles and histidine-tagged proteins secreted into the supernatant of infected High-Five insect cell were purified on Co2+ metal-chelate agarose columns. Eluted products were dialysed overnight in PBS. The yield, integrity and purity of the recombinant proteins were estimated by analysis on SDS gel, comparing to BSA standards, and by western blotting using the anti-V5 antibody. All of the proteins coupled to the Luminex beads were estimated to be at or above 80% purity. The sizes of the different recombinant proteins ranged from 45 to 60 kDa.
Covalent coupling of recombinant PfEMP1 proteins to beads
Carboxylated Luminex beads were covalently coated with the different PfEMP1 protein domains through an interaction of their carboxyl groups and the amino groups on the proteins following the procedure suggested by the manufacturer. Beads (1.25 × 107 beads/ml) were brought to room temperature, vortexed for one minute and transferred to Eppendorf® tubes. The supernatant was removed after centrifugation for one minute at 16,000 × g. 1 ml of distilled water was added to the beads, vortexed to re-suspend followed by centrifugation for one minute at 16,000 × g. The beads were sonicated in a water bath sonicator into suspension and centrifuged for one minute at 16,000 × g. The supernatant was removed by a pipette and 1 ml of activation buffer (0.1 M NaH2PO4, pH 6.2) added to the pellet and vortexed to re-suspend. In separate tubes Sulfo_NHS and EDC were reconstituted to 50 mg/ml and 125 μl of each added to the beads, vortexed and incubated at room temperature for twenty minutes with inversions in the dark. The beads were centrifuged for one minute at 16,000 × g, re-suspended in 1 ml of 50 mM MES pH 5.0, centrifuged for one minute at 16,000 × g and the supernatant removed. The MES wash was repeated. The beads were re-suspended in 500 μl of MES. In separate tubes, the different protein samples (100 μg of each) were mixed with MES to a final volume of 500 μl and each was added to a separate bead population and incubated at room temperature for two hours in the dark with inversions. The beads were centrifuged for one minute at 16,000 × g and the supernatant removed. The beads were washed twice in 1 ml of PBS/TBN (0.02% Tween-20, 0.1% BSA and 0.05% sodium azide in PBS pH 7.4. The beads were re-suspended in 1 ml of PBS/TBN and stored at 4°C in the dark. To determine if coupling was effective, aliquots of the different bead sets were prepared for analysis as described below and analysed on the BioPlex100 system.
Analysis of coupled beads on the BioPlex100 system
The coated beads were diluted 1:333 in Assay Buffer E (ABE buffer: 0.1% BSA, 0.05% Tween-20, 0.05% sodium azide in PBS pH 7.4) and 50 μl aliquots of were dispensed into the wells of a 1.2 μm filter bottom 96-well microtiter plate (MSBVS 1210, Millipore, USA) pre-wetted with ABE buffer. The beads in 96-well plates were washed three times with ABE using a vacuum manifold (Millipore, USA). Frozen plasma samples were thawed at room temperature, mixed by vortexing, and spun at 16,000 × g for five minutes to remove particulates. Plasma samples were diluted 1:80 in ABE buffer and 50 μl aliquots of diluted sample was added to the beads and incubated in the dark on a shaking platform at 1100 rpm for thirty seconds followed by 300 rpm for thirty minutes. Excess antibody was removed using a vacuum manifold followed by three washes in ABE. 25 μl of biotinylated human IgG detection antibody diluted 1:500 in ABE was added to the beads, incubated in the dark with shaking at 1100 rpm for thirty seconds followed by 300 rpm for thirty minutes and washed three times in ABE. 50 μl of phycoerythrin-conjugated streptavidin diluted 1:500 in ABE was added to the beads and incubated in the dark with shaking at 1100 rpm for thirty seconds followed by 300 rpm for ten minutes. Excess phycoerythrin conjugated streptavidin was removed followed by three washes in ABE. The beads were then re-suspended in 125 μl of ABE and analysed on the BioPlex100 system. The reader was set to read a minimum of 100 beads with identical unique detection signal and the results were expressed as median fluorescent intensity (MFI). In order to determine whether the human IgG detection antibody bound non-specifically to the coated beads, the beads were analysed against naïve plasma sample from Danes who have not been exposed to malaria.
Multiplexing and lyophilization of beads
Equal volumes of the coated beads were pooled together and mixed by vortexing. This bead mix was divided in half and one half was stored at 4°C in the dark. Sucrose and Tween 20 were added to the other half to 3% and 0.05% respectively, mixed by vortexing and single-use aliquots were lyophilized (adVantage, Wizad™ 2.0, Virtis) in polypropylene vials, sealed under nitrogen gas and stored at -80°C. Immediately prior to use, lyophilized beads were reconstituted with distilled water and used for analysis as described above.
Enzyme-linked immunosorbent assay (ELISA)
These same samples were analysed by the bead-based assay and the results compared to those from ELISA.
Protein-coupled beads are recognized by hyper-immune plasma
Relationship between ELISA and bead-based assay
Impact of multiplexing on protein-specific reactivity
Stability of protein-coated beads at 4°C
Stability of protein-coated beads after lyophylization
Inter-assay and intra-assay variation for the bead-based assay
Inter-assay and intra-assay variation expressed as coefficient of variation (CV) were determined using the pooled hyper-immune plasma. The inter-assay CV was 13.4% when measured in a single well on ten consecutive plates. The intra-assay CV was 8.0% when measured in eight wells on a single plate.
One of the major challenges faced by researchers dealing with malaria in young children is the limited amount of plasma volume easily obtainable from infants and toddlers. Immunoassays such as ELISA require approximately 1 μl of sample for each analyte. In contrast, the multiplex assay has a major advantage of being able to theoretically analyse one hundred antigens simultaneously in 1 μl of sample. Fouda et al  reported a multiplex assay for measuring antibodies to nine P. falciparum malaria proteins. The multiplex assay described here measures antibody recognition to twenty eight recombinant P. falciparum erythrocyte membrane protein 1 (PfEMP1) proteins simultaneously in 1 μl of plasma sample. To demonstrate effective coupling, the coated beads were analysed individually against pooled hyper-immune plasma (Liberian individuals) and pooled naïve plasma (Danes) and were able to show a significantly higher recognition of the pooled hyper-immune plasma compared to the pooled naïve plasma for each individual coated bead (Figure 1). These results suggest that the recombinant PfEMP1 proteins were effectively coupled onto the beads.
Jacobsen  suggested guidelines for intra and inter-assay coefficients of variation of <10% and <20% respectively for bead-based assays. The bead-based assay reported here has intra- and inter-assay coefficients of variation of 8% and 13.4% respectively, well within the range suggested. Previous studies [7, 14] suggest that the bead-based assay is more sensitive than ELISA in measuring antibody recognition in human plasma samples. For the protein D5, the sensitivity of the bead-based assay seemed to be markedly higher than in the corresponding ELISA assay, but for the three other proteins, the sensitivity of the ELISA assay seemed to be at level with or even a bit higher than the corresponding bead-based assay. The main difference between the results obtained in the two assays was seen in samples with high levels of antibodies, where ELISA readings seemed to have reached a sealed maximal value before the bead-based assay. In a few samples, there was a marked difference between the ELISA and bead-based assay readings (relatively low ELISA value/high bead-based assay values or vise versa). This could be due to the fact that different parts of the recombinant proteins are accessible by antibodies when the proteins are bound to a surface or a sphere. The major advantage of the bead-based assay over ELISA is its ability to measure several analytes simultaneously in a small volume of sample [15, 16]. However, antibody competition and or blocking are to be expected if antigens that share antibody epitopes are detected together in a multiplex. To check for these all twenty eight PfEMP1 coupled beads were analysed in both single and multiplex formats and did not identify significant competition (Figure 3).
Stability of reagents is especially important if results are to be compared over an extended period. It has been shown that coupled beads remained stable for nine months with storage at 4°C . There was a decreasing trend in the MFI values of coupled beads after storage at 4°C in the dark for seven months (Figure 4). However this problem can be resolved by lyophilizing the beads since MFI values for lyophilized beads do not differ significantly from those obtained using non-lyophilized beads (Figure 5). The lyophilization step is a novel approach to maintain the stability and extend the shelf life of coupled beads. This provides a major advantage of reducing batch-to-batch variation when for example analysing plasma samples that are collected over many years in longitudinal studies.
The multiplex assay described here can measure simultaneously antibodies against twenty eight recombinant PfEMP1 proteins in human plasma. The assay also uses less sample volume, is faster and less labour intensive compared to ELISA. The assay shows a novel approach for maintaining the stability and extending the shelf life of coated beads thus reducing batch-to-batch assay variation. Thus, the assay described here provides a useful tool for studying the immunology of malaria especially in young children where the plasma volume available is usually limited.
We thank Susanne Pedersen and Sunthorn Pond-Tor for excellent technical assistance. The study received financial support from the Danish Council for Health Science Research (grant no. 271-05-0427), a grant from the Foundation for the National Institute of Health through the Grand Challenges in Global Health initiative. GKKC is supported by the Gates Malaria Partnership and ATRJ is supported by the Howard Hughes Medical Institute.
- Dobios P, Pereira da Silva L: Towards a vaccine against asexual blood stage infection by Plasmodium falcparum. Res Immunol. 1995, 146 (4-5): 263-275. 10.1016/0923-2494(96)80261-6.View ArticleGoogle Scholar
- Ofori MF, Dodoo D, Staalsoe T, Kutzhals JAL, Koram K, Theander TG, Akanmori BD, Hviid L: Malaria-induced acquisition of antibodies to Plasmodium falciparum variant surface antigens. Infect Immun. 2002, 70: 2982-2988. 10.1128/IAI.70.6.2982-2988.2002.PubMed CentralView ArticlePubMedGoogle Scholar
- Dodoo D, Staalsoe T, Giha H, Kurtzhals JAL, Akanmori BD, Koram K, Dunyo S, Nkrumah FK, Haviid L, Theander TG: Antibodies to variant antigens on the surface of infected erythrocytes are associated with protection from malaria in Ghanian children. Infect Immun. 2001, 69: 3713-3718. 10.1128/IAI.69.6.3713-3718.2001.PubMed CentralView ArticlePubMedGoogle Scholar
- Vignali DAA: Multiplexed particle-based flow cytometric assays. J Immunol Methods. 2000, 243 (1-5): 243-255. 10.1016/S0022-1759(00)00238-6.View ArticleGoogle Scholar
- De Jager W, Prakken BJ, Bijlsma JWJ, Kuis W, Rijkers GT: Improved multiplex immunoassay performance in human plasma and synovial fluid following removal of interfering heterophilic antibodies. J Immunol Methods. 2005, 300 (1-2): 124-135. 10.1016/j.jim.2005.03.009.View ArticleGoogle Scholar
- Opalka D, Lachman CE, MacMullen SA, Jansen KU, Smith JF, Chirmule N, Esser MT: Simultaneous quantification of antibodies to neutralizing epitopes on virus-like particles for human papillomavirus type 6, 11, 16, and 18 by a multiplex Luminex assay. Clin Diagn Lab Immun. 2003, 10: 108-115. 10.1128/CDLI.10.1.108-115.2003.Google Scholar
- Fouda GG, Leke RFG, Long C, Druilhe P, Zhou A, Taylor DW, Johnsen AH: Multiplex assay for simultaneous measurement of antibodies to multiple Plasmodium falciparum antigens. Clin Vaccine Immunol. 2006, 13: 1307-1313. 10.1128/CVI.00183-06.PubMed CentralView ArticlePubMedGoogle Scholar
- Lusingu JPA, Vestergaard LS, Mmbando BP, Drakeley CJ, Jones C, Akida J, Savaeli ZX, Kitua AY, Lemnge MM, Theander TG: Malaria morbidity and immunity among residents of villages with different Plasmodium falciparum transmission intensity in North-Eastern Tanzania. Malar J. 2004, 3: 26-10.1186/1475-2875-3-26.PubMed CentralView ArticlePubMedGoogle Scholar
- Heegaard ED, Qvortrup K, Christensen J: Baculovirus expression of erythrovirus V9 capsids and screening by ELISA: serologic cross-reactivity with erythrovirus B19. J Med Virol. 2002, 66: 246-252. 10.1002/jmv.2137.View ArticlePubMedGoogle Scholar
- Jensen ART, Magistrado P, Sharp S, Joergensen L, Larvtsen T, Chiucchiuini A, Salanti A, Vestergaard LS, Lusingu JP, Hermsen R, Sauerwein R, Christensen J, Neilsen MA, Hviid L, Sutherland C, Staalsoe T, Theander TG: Plasmodium falciparum associated with severe childhood malaria preferentially expresses PfEMP1 encoded by group A var genes. J Exp Med. 2004, 199: 1179-1190. 10.1084/jem.20040274.PubMed CentralView ArticlePubMedGoogle Scholar
- Jensen ATR, Ismail A, Gaafar A, El Aassan AM, Theander TG: Humoral and cellular responses to Sucrose regulated protein 78 – a novel Leishmania donovani antigen. Trop Med Int Health. 2002, 7: 471-476. 10.1046/j.1365-3156.2002.00880.x.View ArticlePubMedGoogle Scholar
- Jensen ART, Gasim S, Ismail A, Gaafar A, Kurtzhals JAL, Kemp M, EL Hassan AM, Kharazmi A, Theander TG: Humoral and cellular immune response to synthetic peptides of the Leishmania donovoni Kinetoplastid membrane protein-11 . Scand J Immuno. 1998, 48: 108-109.View ArticleGoogle Scholar
- Jacobson RH: Validation of serological assays for diagnosis of infectious diseases. Rev Sci Tech. 1998, 17: 469-526.PubMedGoogle Scholar
- Carson RT, Vignali DAA: Simultaneous quantitation of 15 cytokines using a multiplex flow cytometric assay. J Immunol Methods. 1999, 227 (1-2): 41-52. 10.1016/S0022-1759(99)00069-1.View ArticlePubMedGoogle Scholar
- Biagini RE, Murphy DM, Sammons DL, Smith JP, Striley CA, MacKennzie BA: Development of multiplexed fluorescence microbead covalent assays (FMCAs) for pesticide biomonitoring. Bull Environ Contam Toxicol. 2002, 68: 470-477. 10.1007/s001280278.View ArticlePubMedGoogle Scholar
- Fulton RJ, McDade RL, Smith PL, Kienker LJ, Kettman JR: Advanced multiplex analysis with the FlowMatrix™ system. Clinical Chemistry. 1997, 43: 1749-1759.PubMedGoogle Scholar
- Biagini RE, Sammons DL, Smith JP, MacKennzie BA, Striley CAF, Semenova V, Steward-Clark E, Stamey K, Freeman AE, Quinn CP, Snawder JE: Comparison of a multiplex fluorescent covalent microsphere immunoassay and an enzyme-linked immunoassay for measurement of human immunoglobulin G antibodies to anthrax toxins. Clin Vacc Immunol. 2003, 11: 50-55.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.